State investigators have cited a nursing home for neglect in two fatal fall cases that occurred at the Minnesota facility just five months apart – one attributed to a faulty mechanical lift and another to a negligent aide.
The Star Tribune reports the first incident involved a resident who fell while being assisted in the bathroom by a nurse’s aide who reportedly failed to use a gait belt (also known as a transfer belt) on her walker as she made her way to the bathroom. When the woman left the bathroom, she fell and struck her head on a wall, dying from brain hemorrhaging several days later. The aide later explained she didn’t use the belt because she had forgotten it in another resident’s room. An investigation by the state concluded it was the aide’s fault for not properly using the equipment. She was disciplined with a five-day suspension from work and staffers were retrained on why using the gait belt is necessary.
Then a few months later, another resident suffered a fatal fall after slipping from a mechanical lift – one that nursing home staffers knew had a defective part. In that instance, the state did find the mechanical defect was the problem, but cited the nursing home anyway because there was evidence staffers were aware of those problems and used it anyway. The resident had been placed into the lift, but soon after fell onto the floor when one of the safety tabs popped off, resulting in the harness disengaging, dropping her. She suffered a broken leg, but died days later due to complications. An investigative report indicated the facility did not maintain the machine according to the instructions by the manufacturer. In fact, three of the four lifts in use at the facility reportedly had rubber safety tabs that often cracked or loosened, rendering the machines unsafe. The facility reportedly had no procedure through which to monitor this danger to residents. Continue reading →